Fresenius Medical Care Peritoneal Dialysis Connections Baxter to staysafe Connection
Connections using Fresenius Medical Care products This inservice is not intended to replace the advice or training from your physician. Refer to the Instructions For Use (IFU) for detailed device description, instructions, contraindications, warnings, and precautions. Fresenius Medical Care, DEL-CLAMP, Stay●Safe, Safe-Lock, DELFLEX and FREEDOM are trademarks of Fresenius Medical Care Holdings Inc. or its affiliated companies. All other trademarks are the property of their respective owners.
Fresenius Products Baxter is a trademark of Baxter International Inc. or its affiliates. Instructions on how to use Baxter products were reproduced from the manufacturer’s literature and were current at the time of print. Instructions are included for the patient's convenience but Fresenius Medical Care North America does not specifically endorse the use of these products or assume any liability related to their use. Should there be any differences between the instructions listed in this manual and the instructions for use provided with the medical devices referenced, the instructions provided with the medical device should be taken as accurate."
Connections using Fresenius Medical Care products Note: for clarity the model is shown not wearing personal protective equipment (PPE). Follow your facility’s policies and procedures for use of personal protective equipment. It is vital to maintain sterility of the connectors during connection procedures.
Baxter patient to staysafe system A staysafe Extension Set with Luer-Lock patient connector will allow the patient with a Baxter PD system to use staysafesupplies while in the hospital. staysafeExtension Set with Luer-Lock patient connector staysafe System Baxter PD patient
Baxter patient to staysafe system Gather supplies staysafe Extension Set with Luer-Lock patient connector 12 inch (P/N # 050-95004) Or 18 inch (P/N # 050-95005) staysafesolution set and related supplies for transfer set connection and exchange procedure.
ent extension set Baxter patient to staysafe system • Remove the blue sterility protector from the end of the Fresenius staysafe Extension Set . • Remove the mini-cap from the Baxter patient extension set. • 3. Attach the Fresenius staysafe Extension Set to the end of the Baxter extension set. Proceed with the staysafe CAPD or APD exchange. staysafe Extension Set Baxter Extension Set 1 2 3
Titanium luer lock adapter to staysafe system staysafe Extension Set with Luer-Lock patient connector 12 inch (P/N # 050-95004) Or 18 inch (P/N # 050-95005) staysafesolution set and related supplies for transfer set connection and exchange procedure. Remove the blue sterility protector from the end of the Fresenius stay● safe extension set and attach to the luer lock catheter adapter.
DISCHARGE INSTRUCTIONSFOR ALL CONNECTIONS • The patient should return to his/her home or clinic with the additional extension in place. This practice will allow the patient to return home and resume dialysis without interruption. • Instruct the patient to remove any adaptors added to their extension set at the beginning of the next exchange or as directed by the PD clinic nurse.
Choose the type of catheter or transfer set and the type of system utilized. Look down the column and across the row to the adapter needed. Click on the icon in the box to take you to the procedure
Please contact your Fresenius Clinical Sales Representative at 1-800-662-1237 if you have questions regarding the use of a Fresenius PD system or adapter.
CD Available through Customer Service 800-323-5188 (P/N # 100676-01)